Image intensifier is often applied in TV remote X-ray machines, such as fluoroscopy machines and gastrointestinal machines. So it’s important to ensure that Image intensifier works properly.
Image intensifier failure usually occurs in two cases: one is no image at all;The other is that geometric shapes can never be perfectly proportioned to the shapes of the corresponding objects. This phenomenon is often called image distortion. The main reason is that Image intensifier itself has two types of distortion.The other is asymmetry, often called S distortion.
Image intensifier distortion fault is a common fault. When the ground line of video signal is disconnected, there will be a fault like integral distortion and maintenance of the distortion phenomenon.
Usually, the method of first testing power supply is normal, then to determine whether the video signal line ground wire is good, the final test Image intensifier high voltage driving circuit of each test point voltage is normal, if it is a Image intensifier vacuum part distortion caused by aging, is unable to repair, so the distortion of fault factors need to balance and make the correct maintenance plan.
In this regard, Newheek Image intensifier can provide quality detection and maintenance services. Can meet the needs of different customers.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China