The inner liner is the core component of the X-ray image intensifier. Its interior is vacuum designed. Customers often consult the image intensifier liner maintenance. Excessive use or improper operation will cause poor vacuum in the inner tank, which will affect the imaging quality of the image intensifier. When the power is on and no radiation is applied, the output screen will show the half-crescent light. The larger the half-crescent bright area, the inner tank The worse the vacuum is. At this time, we will recommend that the customer do an aging experiment on the image intensifier, which is to provide the image intensifier with long-term power supply for 1-3 days to observe whether this phenomenon still exists. If the vacuum defect is serious, it is recommended that the customer replace the inner tank or image intensifier.
If you need to repair or purchase an X-ray image intensifier, please contact us!
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China