The maintenance engineer W on the call to inquire about the VP-33314 high-voltage power supply, which is equipped with E5804SD-P3 X-ray image intensifier. Manager Wu said that the X-ray image intensifier in the hospital no longer produces images. It is estimated that the power supply is broken. Ask us No repairs.
Tell the customer how to detect the high-voltage power supply, and tell the customer to measure whether it is a power supply problem first. The specific operation method is as follows:
Use DC multimeter to measure (high voltage knob can only be measured with high voltage meter)
1) Test low voltage +24V: if the voltage is normal, the high voltage board is broken; if there is no voltage, the low voltage board is broken
2) Measure the high voltage: the high voltage end is normal and the inner tank is broken; the high voltage end is not normal and the high voltage board is broken.
According to this method, the customer first judges whether the high-voltage power supply is damaged. If the power supply is normal, it means that the inner tank of the X-ray image intensifier is out of order and needs to be mailed to our company for inspection and repair. After consultation with the hospital, Manager Wu decided to mail the X-ray image intensifier to our company for corresponding inspection.
Our company received the X-ray image intensifier mailed by the customer. After a comprehensive inspection of the customer, our engineers found that there was no problem with the high-voltage power supply, but the X-ray image intensifier’s inner tank was aging and caused no image failure. , It is recommended that customers replace the new liner of X-ray image intensifier. Finally, the customer approved our products and after-sales service, and agreed to replace the new liner of the X-ray image intensifier.
All the leaders and engineers of the hospital, if the X-ray image intensifier in your hospital has a similar situation,Welcome to contact us.
Tel: (+86) 17616362243
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China