Some customers come to inquire how to solve the leakage of high-voltage power supply. The function of the high-voltage power supply of the image intensifier is to change the low-voltage current into the high-voltage current through the pulse transformer.
When the high-voltage power supply leaks, when connecting the high-voltage power supply high-voltage line to the corresponding high-voltage output port of the high-voltage power supply according to the wire number, the high-voltage line must be inserted to the end and the high-voltage line must be fastened. If there is still leakage, you can unplug the high-voltage line when the power is off, and apply a high-voltage silicone grease to the line and reconnect it to solve the discharge problem. At the same time, please note that when using the equipment, do not place the image intensifier in a humid environment.
The models of image intensifier high-voltage power supplies are: VP33310, VP33312, VP33314, VP33457, VP33467, etc. If you have an image intensifier high-voltage power supply that needs to be repaired or replaced, you can contact us at any time and we can do a free test.
Tel: (+86) 17616362243
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China