* Health Center purchased newheek NK5761HD-P1P5 X-ray high-voltage power supply with E5804 image intensifier. The image is not very clear because the engineer does not know how to install and debug. In addition, the X-ray image intensifier is installed in the basement. Discharge phenomenon. Similar to this situation, it is recommended that customers mail the entire X-ray image intensifier directly, our company will do a comprehensive inspection for free, and install and debug the X-ray high-voltage power supply.
1. The precautions mentioned below must be carried out on the premise that the inner liner of the X-ray image intensifier can be used normally.
2. Before installation, please clean the high-voltage line connected to the X-ray high-voltage power supply on the X-ray image intensifier with alcohol and dry it to prevent the high-voltage line from being dirty and causing discharge after installation.
3. When installing, connect the cleaned high-voltage line to the corresponding high-voltage output port of the X-ray high-voltage power supply according to the line number. Remember that the high-voltage wire must be inserted to the end when connecting, and the high-voltage wire must be fastened. In order to avoid the loosening of the high-voltage line and sparking.
If the user cannot successfully install the high-voltage power supply according to the X-ray high-voltage power supply manual and installation precautions, it is recommended to mail the X-ray image intensifier to our company. We will install and debug it for free. If you still don’t understand, please contact us.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China