Many repair shops will call to inquire how to repair the output screen of the image intensifier. Let’s take a look at it together.
When the image intensifier is in use, the output screen has a flickering phenomenon, which means that the output screen of the intensifier will flicker and flicker when the intensifier is powered on but no radiation is applied. In this case, consider whether it is correct or not. The image intensifier output screen discharges.
The discharge of the output screen of the booster is mainly caused by the aging of the insulation glue of the output screen of the booster. Therefore, all the insulation glue of the output terminal must be removed, and then the sealing and baking process can be performed to solve the phenomenon of the output screen discharge. . This requires the use of professional packaging equipment, do not do it without authorization, so as not to cause irreparable losses.
If your impact enhancer is in a similar situation, please call for consultation. We will test your machine first, and then issue you the most professional repair plan.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China