The liner of the image intensifier is also called the image intensifier tube. It is the most important part of the image intensifier. Many maintenance companies will call to inquire about the repair of the liner of the image intensifier. Let’s take a look at it together.
When the image intensifier is energized but without radiation, the output screen of the intensifier shows half-crescent light, it is necessary to consider whether the inner tank of the intensifier has poor vacuum. Generally, the larger the bright area of the half-crescent, it means that the vacuum of the inner tank is poor. The more serious the phenomenon.
When this phenomenon occurs in the image intensifier, first, if the inner tank vacuum is not very serious, we will carry out a period of aging test treatment (provide power for the intensifier for a long time for 1-3 days) to observe whether this phenomenon occurs Improved; secondly, if the inner tank vacuum is bad, it cannot be repaired. Generally, it is recommended that the customer replace the inner tank or repurchase the image intensifier.
If your impact enhancer has a similar situation, please come to consult, we will first inspect your machine, and then issue the most professional maintenance plan for you.
Tel: (+86) 18953613955
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China