Recently, our company’s engineers discovered a phenomenon in the maintenance of image intensifier tube equipment: when the bed was getting up and lying down, the image on the display showed that there were fluid objects flowing from the edge or from the middle to the edge, and finally the display turned white. As a result, the image cannot be seen clearly. (This phenomenon often occurs when the G3 of Thales tube falls off).
The reason for this phenomenon is that the electrode inside the image intensifier tube falls off. After many tests and communication with the merchant, it was found that this phenomenon could not be repaired. You can only replace it with a new inner tank or purchase a new image intensifier tube to replace it. The inability of the shadow to focus is also a common problem. So what causes it?
After disassembling the machine, there are two situations where this fault appears the most:
1. The high probability is caused by high voltage discharge, which needs to be further judged according to the specific model.
2. There is a bad vacuum in the inner tank.
We, Newheek, are a professional manufacturer of X-ray machine accessories and equipment. Its products are exported to dozens of countries at home and abroad. Its products mainly include X-ray machine hand switch, medical filming bed, high-voltage cable, image intensifier tube, high-voltage power supply Wait. If you are interested in our products, you can contact us.
Tel: (+86) 17616362243
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China