Today, I will give you some knowledge. What are the faults of the X-ray image intensifier? Let me talk about the first one. If there is no image, there are three situations: 1. The inner tank of the X-ray image intensifier is damaged, 2. X-ray The high-voltage power supply of the image intensifier is poorly connected. This can be tested. If the power supply of the 3X-ray image intensifier fails, if the power supply is broken, you have to find a replacement. This is also easy to say, if it is a long time, it will be difficult to find , We have received such a customer here, the high-voltage power supply is broken, and there are not many places, but we have one, which is a long time away. We found this place through other channels, and we sold it to him at the market price. If he meets other manufacturers and distributors, he may ask for more money.
Closer to home, the second one is that the image is thin. There are two cases of this kind of situation. 1. The enhancement tube of the X-ray image intensifier is aging, and this one needs to be replaced. 2. The high pressure of the X-ray image intensifier is low. This can be specific. Check it out.
In the third case, the image is blurred, and there are three situations that cause this. 1. The vacuum degree of the intensifier tube of the X-ray image intensifier is poor. 2. The focus voltage of the intensifier tube of the X-ray image intensifier is inappropriate. This has to be tested by professionals. 3. The output screen has a lot of dust. You have to wipe this by yourself. Be careful not to get water in, and then the image flickers. 1. The vacuum of the intensifier tube of the X-ray image intensifier is poor, and 2. Enhancement Although there is high-voltage discharge interference, you have to find a special engineer to check it. The above situations are basically the most common. Does everyone understand?
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China