Today I will explain to you the X-ray high-voltage power supply that powers the image intensifier.
Not long ago, the X-ray high-voltage power supply of an image intensifier in a hospital ignited when it was turned on, and asked whether this situation could be repaired. In view of this situation, because we are not at the scene and have not seen the malfunction, if the customer wants to repair it, it is recommended to mail it directly. Our company is responsible for detecting the cause of the failure free of charge, and judging whether it can be repaired based on the result. At present, our company has received the X-ray high-voltage power supply mailed by the customer. After the inspection by our engineers, the high-voltage end has been carbonized and cannot be repaired. The customer asked: Can’t you change the high-voltage end? Tell the customer that because the internal burn-through, even if the high-voltage end is changed, it will not help. The customer asked again: What caused the internal burn through? It was reported that because the X-ray high-voltage power supply had a slight discharge and was not treated in time, it was carbonized and burned through. In this case, if you continue to use it, the inner liner is also easy to burn out. The customer says it understands.
Dear users, if your equipment has this kind of problem, remember to repair it in time. The best way is to replace the X-ray high-voltage power supply in time. If you have any needs, please feel free to contact us. After receiving the information, we will contact you as soon as possible.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China