I received a call from an industrial customer repair without images X-ray image intensifiers.At present, the customer uses TH59464 X-ray image intensifiers. The current problem is that the X-ray image intensifier has no image. The customer tested that the high-voltage power supply on the X-ray image intensifier was faulty, but they would not replace it themselves, because professional X-ray image intensifier repairs should be repaired by professionals who produce X-ray image intensifiers. They use X-ray image intensifiers to detect the welding of gold wires in the product. Customers If you plan to mail the X-ray image intensifier, our company can detect the cause of the X-ray image intensifier failure free of charge. If it is determined that the high-voltage power supply is broken, you can directly replace the high-voltage power supply and send the X-ray image intensifier back. The customer has been mailed to SF Express. The high-voltage power supply price is *yuan. The round-trip mailing needs to be borne by the customer.
Our company received the X-ray image intensifier. After testing the inner tank of the X-ray image intensifier, the high-voltage power supply burned out. After we replaced it with a new high-voltage power supply, the image was displayed normally. The results of the test made by our company with the ground wire and the circuit board with 70KV-0.5MA, the customer is very satisfied with the image effect. Said that they generally use 100KV-15UA for testing on site. The final customer pays, our company packs and ships to the customer’s designated address.
Today’s sharing is here first. In addition, our company specializes in the production of various vertical and mobile chest film racks, various types of X-ray machine parts, please call to discuss!
Tel: (+86) 17616362243
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China