A customer from a health clinic recently ask whether the VP-33314 X-ray high-voltage power supply can be repaired? The specific communication content with customers is as follows:
The X-ray high-voltage power supply of their E5804HD-P3 image intensifier is burned out. Can your company repair it? Let the clinic take a picture and send it to me later. Please add my WeChat. Tell the customer that the label parameters of the X-ray high-voltage power supply are generally on both sides. According to the model of the image intensifier you provide, it should be a VP-3331* X-ray high-voltage power supply. Let the clinic check it.
At present, the customer has taken pictures. This X-ray high-voltage power supply is VP-33314. At present, our company has replacement parts. As for the repair, our company can only confirm after inspection. If the clinic is sure to repair it, you can send it to us for inspection first. If the damage is not serious, we try to repair it.
I have confirmed with the clinic, and the clinic agrees to mail it. Please send me the address.
At present, our company has received the X-ray high voltage power supply. From the outside of the power supply, the circuit board at the semiconductor triode is burned through. From the traces, the X-ray high-voltage power supply has been repaired once. If it is repaired again, it will still be burned after a while, so our recommendation is no The value of the restoration. Finally, the customer directly purchases the new X-ray high-voltage power supply.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China