Many customers often ask in the process of consulting high-voltage power supply products, how to repair the abnormal discharge of high-voltage power supply, when it comes to maintenance, the first thing we think of is how to maintain and maintain, and what to pay attention to during use, so as to avoid abnormal discharge. Let’s take a look at the precautions for using high-voltage power supplies.
(1) Under the premise that the liner of the image intensifier can be used normally (except for the damage of the liner, such as discharge, leakage, air intake, pressing down the input screen, etc.).
(2) Before installation, please use alcohol to clean and dry the high-voltage line connected to the high-voltage power supply on the image intensifier to prevent the high-voltage line from being dirty and discharge after installation.
(3) When installing, connect the cleaned high-voltage line to the corresponding high-voltage output port of the high-voltage power supply according to the line number. Remember that when connecting, the high-voltage line must be inserted into the bottom, and the high-voltage line must be tightened. So as to prevent the high-voltage wire from loosening and sparking.
(4) Whether the high-voltage line is cleaned and tightened at the same time, but there is still a slight discharge at the high-voltage end. In the case of a power failure, the high-voltage line can be pulled out, and the discharge problem can be solved after applying high-voltage silicone grease on the line and reconnecting it.
(5) In the process of adjusting the image, it can be adjusted according to the method mentioned in the manual.
(6) Strictly require the use environment of the equipment, and fundamentally prevent the emission and combustion of the equipment caused by the abnormal working environment such as air humidity.
We Weifang ewwheel Electronic Technology Co., Ltd. is an import and export trading company that produces x-ray machines and accessories, please call for more details about high-voltage power supplies.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China