Image intensifier This situation is caused by magnetic field interference, geomagnetism. Geomagnetism exists everywhere, and we do not have a degaussing wire on our influence booster, so this happens. (Because this shield is expensive, it is usually not used)
If there is a degaussing coil on the old shadow increase in front of the customer, you can tell the customer to remove the degaussing coil on the shadow increase and install it on our shadow increase.
Let the client put the image in the center and observe
Remove the large flange (input flange) of the original image intensifier and you will see a thin shield. Some are on the input flange, and some are on the input end of the shadow booster liner. After finding it, copy it down and put it on our booster.
Tel: (+86) 17616362243
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China