Problems with the high voltage power supply of the image intensifier will affect the normal operation of the image intensifier. In fact, daily maintenance is very simple.
High voltage power supply is an important part of image intensifier.
When there is a problem with the high voltage power supply of the image intensifier, we need to repair it.
What should we pay attention to?
The following precautions must be taken (in addition to tank damage, such as tank leakage, air intake, etc.) while ensuring that the tank of the image intensifier can be used normally.
Before installation, clean the high voltage line connected to the high voltage power supply on the image intensifier with alcohol and dry it to prevent the high voltage line from being dirty and causing discharge after installation.
When installing, please connect the cleaned hV wire to the corresponding HV output port of the HV power supply according to the wiring number.
When connecting, be sure to insert the high-tension cable into the end and tighten the cable.
To avoid loose high voltage cables and sparks.
If the high voltage cable is also cleaned;
It’s also fixed, but there’s still a small discharge at the high voltage end.
The discharge problem can be solved by unplugging the high-voltage cable when the power is turned off and applying some high-voltage silicone grease on the cable and reconnecting it.
Newheek specializes in the manufacture of X-ray machines and radiology equipment. If you are interested in the high voltage power supply used in this type of image intensifier, please contact us.