Side of the image intensifier:
The inner layer of the image intensifier is a glass liner (ie, reinforced tube), and the outside is covered with black insulation to cover the light and is a key component to achieve the function of the image intensifier; the outer layer is the lead layer, which is mainly used to absorb and isolate X-rays that do not produce photoelectric effects. ; Outside is the shield layer of Permalloy (not shown), the main role is to prevent external magnetic fields; the outer layer is a metal shell (ie, the sleeve), which mainly plays a protective and fixing role.
Input terminal of the image intensifier:
Input flange (input interface ring), made of aluminum, fixing the input end and connecting the parts of the diagnostic bed;
Filter plate, made of aluminum (ultra-thin, easily scratched and deformed), can be replaced with a wire grid, but the effect is not as good as the wire grid
Output of the image intensifier:
Output flange (output interface ring), made of aluminum, fixed output end, connected to camera parts.
Newheek specializes in the production and sales of X-ray machines and imaging equipment, can replace Thales, imaging, Toshiba and other equipment, if you are interested in this equipment, welcome to contact us!
Tel: (+86) 18953613955
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China