We have learned about the phenomenon and solutions of the leakage of the output screen of the image intensifier before. Today we will look at the phenomenon and solutions of the leakage of the inner tank of the image intensifier.
In the image enhancement period, the leakage of the gallbladder is shown as follows: the image is blurred, and the voltage at the high voltage terminal drops linearly.
So how to solve this problem? If there is only slight leakage, the output screen can be resealed.
If it is serious, it means that the tank is discharged and needs to be replaced or new products purchased.
The above is the phenomenon and solution of the leakage of the image intensifier tank. If you have any questions or need to buy our products, please consult. Tel（whatsapp）:+8617616362242
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China