Ximage intensifier consists of an image intensifier tube, tube container, optical system and power supply.There is a ray display without image. Take the back cover of the intensifier apart and observe from the camera. When the green light is not visible at the time of exposure, the fault occurs in the image intensifier.
If the primary voltage of the high-voltage transformer is normal, check whether there is high voltage in the anode of the enhancer and the secondary voltage of the high-voltage transformer and whether the connection of the high-voltage line is in good contact.
If a piece of cotton yarn is put on the high-voltage wire, when electrified, the cotton yarn will stand up, indicating that there is a high pressure;If the conditions permit can also be used to the ground whether there is a discharge spark method to prove that the high pressure to the ground by far slowly close, the general distance is about 5cm.
Newheek can not only provide the ximage intensifiers for different sizes, but also get a repair intensifier.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China