High voltage power supply is an important part of x ray image intensifier. What should we pay attention to when the high voltage power supply of x ray image intensifier is in trouble?
The following precautions must be taken to ensure that the inner gallbladder of the x ray image intensifier can be used normally (except for internal gallbladder discharge, leakage, air intake, input screen depression and other internal gallbladder damage). Before installation, please clean and dry the high-voltage wires connected with the high-voltage power supply on the x ray image intensifier with alcohol, so as to avoid the unclean high-voltage wires and cause discharges after installation. During installation, the clean high-voltage wire is connected to the corresponding high-voltage output port of the high-voltage power supply according to the line number. Keep in mind that the high-voltage wire must be inserted to the end and tightened when connecting. In order to avoid the phenomenon of ignition caused by the loosening of the high-voltage wire. If the high voltage wire is cleaned and tightened, there is still a slight discharge at the high voltage end. The high voltage wire can be pulled out in case of power failure, and the discharge problem will be solved when the high voltage silicone grease is re-connected on line.
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China