Common Fault Analysis and Troubleshooting Methods of X-ray Image Intensifier.No image First check whether the high-voltage circuit is faulty, and use a multimeter to measure the output voltage of the high-voltage circuit. If the high pressure is normal, check whether the intensifier tube is damaged. If the image focus is not good, first adjust the focus potentiometer to observe whether the image can be well focused; if it cannot focus well, it can be judged that the voltage of each electrode is wrong, and the focusing circuit and the voltage value of each electrode should be checked; if the voltage value of each electrode is normal, the fault still exists, explain The reinforcement tube needs to be replaced. There are speckles in the image, and there are impurities on the input or output screen of the intensifier tube, indicating that the quality of the intensifier tube is defective, or the intensifier tube is aging. Most of the customers are some medical equipment companies, and they want to ask the hospital if they can repair their image intensifiers.
If you are interested in our image intensifier, welcome to contact us, phone (whatsapp): +8617616362243!
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China