Maintenance engineer Mr. Shi called to inquire VP-33314 high voltage power supply, which is equipped with E5804SD-P3 X-ray machine image intensifier.The manager said the hospital’s X-ray machine image enhancer was no longer working.It is estimated that the power supply is off and cannot be used normally. We hope you can give us valuable advice on the maintenance.
Inform the customer how to detect the high voltage power supply, and tell the customer to first measure if it is a power problem.The specific operation method is as follows:
Measurement using a DC multimeter:
1) Detection of X-ray machine image intensifier low voltage +24V: normal voltage, high pressure plate damage;If there is no voltage, the low pressure plate will fail
2) Measure image intensifier of high pressure X-ray machine: high pressure end is normal, internal groove is broken;The high pressure end is abnormal and the high pressure plate is broken.
According to this method, the user first determines whether the high voltage power supply is damaged.If the power supply is normal, it means that the internal tank of the X-ray machine image intensifier is out of order, which needs to be mailed to our company for inspection and maintenance.After consultation with the hospital, Manager Wu decided to mail the X-ray machine image intensifier to our company for corresponding inspection.
Newheek is a professional manufacturer of X-ray machines and spare parts. If you are interested in this kind of products, please contact us.
Tel: (+86) 18953613955
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China