X-ray machine high-voltage power supply can be said to be the most important part of the image intensifier. So how to test the voltage value of X-ray high voltage power supply?
Before testing the X-ray high voltage power supply, the tools must be prepared first. What tools do you need? what are they doing? The first high-voltage electrostatic machine was the so-called high-voltage instrument. Its function is to measure X-ray high voltage power supply. If the value of the X-ray high voltage power supply is too large, the multimeter cannot be used. You can only use high voltage gauges. The next one is to prepare for the test. For the X-ray high voltage power supply, the last one is the multimeter. If the value is not very large, use a multimeter.
Connect the black wire to case D and the red wire to 1234 for measurement.
It can then be resized. For example, the G3 terminal modulates the voltmeter to the highest or lowest level to detect whether the high-voltage power supply is out of the controllable range.
The preparations are ready, and the most important step is how to check, that is, the inspection process of the X-ray high-voltage power supply: first connect the X-ray high-voltage power supply to the high-voltage instrument and test it. wire, then connect the high voltage gauge to the power source. The first step is to connect the GND terminal of the high voltage meter to the shell of the X-ray high-voltage power supply, turn on the X-ray high-voltage power supply 24V, and finally record the data. One thing that has to be noted is that the high voltage gauge should use a thick wire when measuring HV and a thin wire when measuring G1234. This is the easiest place to go wrong.
Finally, when the high-voltage meter is used up, the X-ray high-voltage power supply must be cut off to avoid unnecessary losses. I think everyone has some understanding of this.
Newheek mainly produces X-ray high-voltage power supplies, image intensifiers, monitors, etc. The products are sold at home and abroad, and have won unanimous praise. If you have needs in this area, please call to discuss: +8617616362243!
Tel: (+86) 18953613955
Whatsapp: +86 18953613955
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China