A customer after-sales service department consults how to measure the high-voltage power supply of the image intensifier, and the response requires a DC multimeter to measure. One of the common failures of image intensifiers is a problem with the high-voltage power supply of the image intensifier.
How to measure the high-voltage power supply of the image intensifier, the specific operation method is as follows:
1. First use a DC multimeter to check the low-voltage end of the image intensifier: if the voltage is normal, the high-voltage board is damaged; if there is no voltage, the low-voltage board is broken.
2. Re-measure the high-pressure end of the image intensifier: if the high-pressure end is normal, the inner tank is broken; if the high-pressure end is abnormal, the high-pressure plate is broken.
How to measure the high-voltage power supply of the image intensifier is a very professional operation. Unless it is an engineer in the after-sales maintenance department, we generally recommend that customers send the intensifier to us for professional inspection.
Tel: (+86) 17616362243
Fax: +86 536 8882360
Company: Weifang Newheek Electronic Technology Co., Ltd.
ADD: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China