What should I do if the image inductor is peeled off? First, determine whether the image intensifier electrode is off, mainly depends on the following aspects:
1 performance phenomenon: the diagnostic bed with image intensifier in the process of lying up, the image on the display shows that the flow of the object flows from the edge part or from the middle to the edge, and finally the display turns white, resulting in invisible image . (This phenomenon often occurs on Thales tubes, mainly as the G3 electrode falls off.)
Then, after judging it, how to solve it?
2 Solution: Unable to repair, replace the liner or purchase new products. If you replace the liner, you can contact NEWHEEK to replace it. Since the replacement of the liner requires professional equipment debugging, the original image must be shipped to the Huarui Image NEWHEEK.